1. Know your application
If you face the challenge of setting up a vibration measurement system for a rotating machine, you need to know the mechanical design and bearing of the shaft. There is a significant difference between measuring vibration of ball bearings and plain bearings. Ball bearings use accelerometers, while plain bearings use proximity probes or both.
2. Environmental and surface temperature
Keep in mind that not all types of accelerometers can be used in very high or very low surface or ambient temperatures. There are accelerometers with built-in electronic components in their sensor casings and sensors without electronic components. Sensors with integrated electronics, such as MEMS and IEPE accelerometers can be used up to 120°C / 248°F (MEMS) and 150°C to 200°C (302°F to 392°F) (IEPE) surface or ambient temperature. For high temperature applications, such as measurements on gas turbines, pipes, and vessels, piezoelectric (PE) sensors must be used.
3. Vibration or shock?
Standard accelerometers are suitable for measuring random vibration on machines, structures, and other components. However, for shock measurements, e.g. in crash tests, very high accelerations occur, so that special accelerometers are required. For these types of tests, use PE accelerometers for a broad measurement range up to 50,000g or MEMS sensors based on piezoresistive sensing elements.
4. Measuring in high-frequency ranges down to DC
Depending on the application, the frequency response of the sensor is essential for measuring every frequency of interest. When measuring structures such as bridges or towers (e.g., wind turbines and modal analysis) the frequency of the vibration measured can be very low, sometimes below 0.5Hz or down to DC. For these applications, you should use sensors with a frequency response down to DC. MEMS-based accelerometers are suitable for the job.
5. Sensor mounting and frequency response
In applications such as machine monitoring, the vibration must be measured over a broad frequency range. Rotating machines with 3000 rotations per minute or higher and roller bearings have frequencies of interest exceeding 3-5kHz. To view the harmonics of these signals, the frequency response of the sensor(s) used must be higher than 3-5kHz. The sensor’s maximum frequency response depends on its fixation on the surface of the measured object. For signal frequencies higher than 3kHz, magnets can be used. However, for the best measurement results, especially for frequencies exceeding 10kHz, the sensors must be fixed with adhesives or mounting studs.
6. Routing and fixing the sensor cable
To measure the vibration of an object, you want to mount the sensor on it, but not have induced vibrations from other parts. Therefore, the sensor cable must also be laid with great care with fixation at several points. With free-flying cables, additional vibration can be introduced by components other than the object measured. This introduced vibration affects the sensor and distorts the signal you wish to measure. Especially when using Piezoelectric sensors, the movement of the cable from the sensor to the charge amplifier can cause charge changes to be transferred with this cable.
7. Avoiding noise in the measured signal
If you see noise in your measurement signal, especially at 50/60Hz, you most likely have a ground loop. Ground loops can transmit interference to the measurement signal as noise, most common at a line frequency of 50/60Hz. Ground loops develop when a common line (e.g., signal return/shield of an IEPE accelerometer installation) is grounded at two points of differing electric potential. To avoid such ground loops, connect the cable screen only at a place to ground potential and use electrical isolated sensing elements when possible.
8. How does the time constant of a charge amplifier affect measurements?
There is a fundamental difference between the two operating modes of charge amplifiers for quasi-static and dynamic measurements. Most charge amplifiers support both measurement types. But before setting up a vibration measurement chain, a clear understanding of the desired measurement is essential to select the right charge amplifier operation mode. For example, measurements in exceptionally low-frequency ranges <0.5Hz should be performed with the quasi-static mode. This is sometimes the case when measuring low frequency pressure fluctuations in high-temperature applications. Typical vibration measurements have a higher frequency content in the signal; therefore the dynamic mode is used.
The time constant of the charge amplifier defines the operating mode for static or dynamic measurements.
9. What is the maximum cable length for an IEPE sensor?
IEPE sensors are powered by a constant current and a DC voltage. The AC signal of the measured acceleration is superimposed on the supply voltage. As a rule of thumb, the higher the constant current, the greater the cable length can be. But keep in mind: maximum cable length depends also on cable quality. Most signal conditioners and I/O-modules for IEPE sensors provide a 4mA constant current that forms a very good optimum between noise immunity, cable length, signal frequency, and current consumption. Measurements with cables longer than 80 m or even 100 m are possible.
10. Does the supply voltage for a MEMS-based accelerometer matter?
The short answer is yes, the supply voltage does matter. Most MEMS sensors have a built-in voltage regulator but are very sensitive to the quality of the supplied input voltage. The quality of the voltage source must be high, should be regulated within +/-0.1%, and the noise and ripple must not exceed 1.5mV RMS. I/O modules designed specifically for MEMS-based sensors provide a stable power supply, and when using these devices, the supply voltage does not cause headaches.
This is how to select the appropriate accelerometer
Download our free whitepaper Vibration Monitoring Essentials for more detailed technical information about acceleration measurement, including the pros and cons for each of the three absolute vibration transducers PE, IEPE, MEMS used for measuring acceleration.
8 Tips to Consider When Buying a Strain Gage Amplifier
The strain gage is one of the most important tools of the electrical measurement technique applied to the measurement of mechanical quantities. As their name indicates, they are used for the measurement of strain.Read more...
Experience a New Level of Possibilities With Our New Website!
We are thrilled to announce the launch of the new Gantner Instrumentswebsite and introduce you to a few of its features.Here’s a sneak peek of what you can expect to find. “Intuitive Data Acquisition Solutions for Modern Testing and Monitoring Applications" reflects our commitment to providing data acquisition solutions with the best performance, adaptability, and […]Read more...
Gantner Instruments Sponsors STEM Education
The world is changing around us. Advances in technology impact everything, especially the world of work. It’s predicted that future workers will spend more than twice as much time on job tasks requiring science, math, and critical thinking than today. At Gantner Instruments, we recognize the importance of STEM education: it holds the keys to success for kids facing a world where STEM knowledge is imperative. For that reason, we are proud to be a Platinum Sponsor of Students for the Exploration and Development of Space at the University of California, San Diego (SEDS UCSD).Read more...
Advanced System Monitoring & Analytics for Smart Grids
Gantner Instruments, in partnership with The University of Cyprus, researches the controllability of renewable energy sources in local self-operated Smart Micro-Grid using innovative data streaming solutions including Crate.io.Read more...